TIANCHI Cow Ultrasound Scanner Machine TC-301 Manufacturer in RO
Animal B ultrasound is a kind of ultrasonic imaging method,which uses B ultrasound probe to transmit ultrasonic waves,at the same time receiving the ultrasonic that reflected by various tissues and displayed on a specific display to form an image.The animal b ultrasound producted by Tianchi has obvious imaging,and there is no excessive clutter interference,people can use it without special operation training,it is convenient and accurate.
Installation and disassembly
Connection between probe and main unit
Probe jack lies in the top of the right side, insert the probe into the jack correctly and turn it clockwise and wound it. There is only one probe jack which can be connected to all the supported optional probes.
Avoid by all means unplugging or plugging the probe connector at state of log on in case the probe and main unit be damaged. Once the probe is connected with the main unit, do not unplug nor plug it at discretion in case poor contact happen.
Must not touch the contact pin of the probe connector.
The probe should be protected from felling off or crashing and the manufacturer assumes no responsibility for this kind of hazard.
Please handle the device carefully.
1) Main unit maintenance
(1)Instrumentation environment should be according to "2.1 operation environmental requirements".
(2)If device enclosure needs cleaning, shutdown the device first and then wipe with alcohol sponges.
(3)Device should not be turn on and off frequently.
(4)When the device does not work for a long time, pack the device according to the instructions on the packing. Store it properly in the warehouse. The storage environment should accord to transportation and storage environmental requirements.
2) Probe maintenance
(1) Probe is an expensive part but frangible. Never hit it or drop it on floor. Please use medical ultrasound Gel during diagnoses. Do daily inspection on the probe enclosure to see whether it is cracked and to protect it from the water inrush.
(2) Check the probe regularly to see whether it is filled with clean medical castor oil, air in it will influence the image quality. If air bubbles appear, turn the probe oil filler hole up and screw off the sealing screw, turn the probe slowly to drive the air bubbles to the screw hole, infuse a little bit of castor oil into the screw hole with syringe with a needle, vent the air bubbles and fasten the sealing screw and finally clean the oil stain on the probe surface.
Henan tianchi is a professional manufacturer of B ultrasound machines.We developed full digital doppler ultrasound diagnosis system independently.The company has completely independent intellectual property rights, unparalleled ultrasound imaging technology.In addition, we also provide free warranty for two years (except for man-made faults), and provide life-long maintenance services. For details, please contact +8618903712477 /+ 8618037138109.
|Minimum Order||1 Piece/Pieces|
|Price for Minimum Order||1 USD|
|FOB price||100.00 USD Piece/Pieces|
|Payment & Price Terms||Delivery within 5 days after payment|
|Delivery Lead Time||Delivery within 1-7 days|
Mr. Jian Yang
TIANCHI Cow B Ultrasound TC-301 in China|